10.4 Geometric Distortion in ACS
The ACS detectors exhibit more distortion than previous HST instruments. The principal reason is that the optics have been designed with a minimum number of components, consistent with correcting for the spherical aberration induced by the Optical Telescope Assembly (OTA), without introducing coma. The result is a high throughput, but with focal surfaces far from normal to the principal rays. The WFC detector is tilted at 22° giving an elongation of 8% while the HRC and SBC have a 25° tilt leading to an elongation of 12%. In each case, the scales in arcseconds per pixel are smaller along the radial direction of the OTA field of view than along the tangential direction.
The orientations of the ACS detector edges are approximately in line with the V2 and V3 coordinate axes of the telescope. Consequently, the eigenaxes of the scale transformation are along the diagonals for WFC, and the apertures and pixels appear nonrectangular in the sky projection. For the HRC and SBC, the situation is even more irregular because the aperture diagonals do not lie along a radius of the HST field of view. Figure 7.9 shows the ACS apertures in the telescope's (V2, V3) reference frame. For a telescope roll angle of zero, this would correspond to an onsky view with the V3 axis aligned with North and the V2 with East.
There is not only a strong geometric distortion of ACS detectors, but a significant variation of the scale across each detector. For the WFC, the scale changes by approximately 10% from corner to corner. For the HRC and SBC, this variation is only about 1% as they cover much smaller fields of view. The Science Instrument Aperture File (SIAF) provides the most accurate values of the scale for all instruments on board HST. The area on the sky covered by a WFC pixel varies by about 18% from corner to corner, corrections for which must be made in photometry of extended objects. This variation of scale creates a problematic effect in combining ACS images by the fact that an integral pixel shift near the center of the detector will translate into a nonintegral displacement for pixels near the edges. Therefore, image alignment and combining require an accurate geometric distortion model.
The HRC geometric distortion calibration is fully described in ACS ISR 200415. The SBC geometric distortion is described in ACS ISR 200802. The WFC geometric distortion is described in ACS ISR 201506. The geometric distortion model is expressed by highorder polynomials for each of the ACS detectors in the form of reference files called as Instrument Distortion Coefficients Tables (IDCTAB) in the FITS format. On the top of the geometric distortion described by the polynomial equations, there are additional filterdependent components of the distortion that are presented by a 2D array and are applied to each row/columns of the WFC or HRC images. This filterdependent component of the distortion is applied via a calibration file called NPOLFILE. In the case of the WFC, there is an additional component of the distortion called pixelgrid irregularities. These are caused by irregularities in the manufacturing process and are also presented by 2D array to correct each row/column on WFC images. This correction is applied via the FITS formatted D2IMFILE. All distortion reference files are installed in the ACS data pipeline and can be used with the Drizzlepac software distributed by STScI.
Additionally, an area of the open cluster NGC188, for which accurate astrometry is available, was used to establish the exact location and orientation of the aperture in telescope coordinates (ACS ISR 200707). At the same time, the scale factors were confirmed.
10.4.1 WFC
The rhombus shape of the WFC is evident in Figure 7.9. The angle between the X and Y axes is 84.9° for WFC1 and 86.1° for WFC2. The geometric distortion map for WFC1 and WFC2 is illustrated in Figure 10.111. A vector diagram shows the contribution of the nonlinear part of a quadratic fit only. At the center of chip WFC1, the scale in the X direction is 0.0493 arcsecond/pixel, and 0.0486 arcsecond/pixel in the Y direction. In the case of WFC2, the scale is 0.0498 arcsecond/pixel in the X direction, and 0.0503 arcsecond/pixel in the Y direction. Between the corner of WFC nearest to the V1 axis and the diagonally opposite corner, the scale increases by 10%. Therefore, WFC1 forms a slightly distorted rectangle 201 × 100 arcsecond^{2} in size, while WFC2 is 203 × 103 arcsecond^{2}. There is a 2.5 arcsecond gap between the two chips. See the SIAF website for more details.
ACS ISR 200708 shows that the linear terms of the WFC geometric distortion are changing over time and distortion corrected positions could be off by up to 0.3 pixels when compared to the beginning of the instrument's lifetime. The scale of the X axis changed after SM4, but the reason for the change is not yet understood. The timedependent distortion has been calibrated for pre and postSM4 (ACS ISR 201506) and its parameters are implemented via the IDCTAB, which is used by both the ACS data pipeline and the Drizzlepac software.
Geometric distortion affects not only the astrometry, but also the photometry, since it induces an apparent variation in surface brightness across the field of view. In order to preserve the photometric accuracy of nondrizzled images, an additional correction to the photometry is required via multiplication of the ACS/WFC flatfielded images by the pixel area map. The effective area of each pixel is shown in Figure 10.112 as a contour plot. The range of area goes from 0.89 to 1.08 times the central value.
10.4.2 HRC
HRC has been unavailable since January 2007. Information regarding the HRC is provided for archival purposes only.
The High Resolution Channel has its edges aligned approximately along the V2 and V3 axes. In this case, the center of the aperture lies on a line passing through the V2V3 origin and making an angle of 22° with the V3 axis. The diagonal of the aperture does not correspond to a radius of the HST field of view, so the distortion has no particular symmetry with respect to the detector axes. The focal plane of HRC is also 25° away from the plane normal to the light path, hence the scales along the axes differ by 14%. The full field of view of the HRC is less than 30 arcseconds, therefore the scale variation over the field is much less than for the WFC, about 1%. At the center, the X and Y scales are 0.0284 and 0.0248 arcsecond/pixel respectively. The average scales across the middle of the detector are 0.02842 and 0.02485 arcsecond/pixel, making the X and Y widths 29.1 and 25.4 arcseconds. The slightly nonsquare projected aperture shape is evident in Figure 7.8. The angle between the X and Y axes on the sky is 84.2°. The geometric distortion map for HRC is given in Figure 10.113, where the residuals from the nonlinearity are scaled by a factor of 10, with residuals as large as 0.14 arcseconds or 4.9 pixels.
As in the case for the WFC, geometric distortion affects not only the astrometry but also the photometry, and a correction for the pixel area is required in order to restore the proper total counts of the target. The effective area of each pixel is shown as a contour plot in Figure 10.114. The maximum deviation from the central value is about 3%. Because the pixel area map is normalized to square pixels of 0.025 arcsecond on a side rather than an area equal to that of the central HRC pixel, the pixel area map ranges from about 1.1 to 1.15.
10.4.3 SBC
The Solar Blind Channel contains the MAMA detector, which is centered close to the HRC position in the (V2, V3) plane and has a slightly larger field of view, about ~35 × 31 arcsec^{2}. The scales and distortions have been directly measured. The maximum distortion displacement is about 2 pixels or 0.06 arcseconds. Figure 10.115 shows the distortion map for the SBC detector.
The HRC and SBC both have much smaller fields of view than the WFC. For the SBC, the X direction scale is 0.0338 arcsecond/pixel and the Y direction scale is 0.0301 arcsecond/pixel. Similar to the HRC, the SBC exhibits a 13% difference between X and Y scales with a variation across the aperture of a little over 2%. As with the other cameras, geometric distortion affects photometry by a variation in pixel size across the SBC.
The map of the effective pixel areas of the SBC is shown in Figure 10.116. Because the pixel area map is normalized to square pixels which are 0.025 arcseconds on a side rather than an area equal to that of the central SBC pixel, the pixel area map ranges from about 1.58 to 1.65. (Meurer et al. 2002). The maximum deviation from the central value is about 5% (ACS ISR 200709 and ACS ISR 200802).

ACS Instrument Handbook
 • Acknowledgments
 • Chapter 1: Introduction
 Chapter 2: Considerations and Changes After SM4
 Chapter 3: ACS Capabilities, Design and Operations
 Chapter 4: Detector Performance
 Chapter 5: Imaging
 Chapter 6: Polarimetry, Coronagraphy, Prism and Grism Spectroscopy

Chapter 7: Observing Techniques
 • 7.1 Designing an ACS Observing Proposal
 • 7.2 SBC Bright Object Protection
 • 7.3 Operating Modes
 • 7.4 Patterns and Dithering
 • 7.5 A Road Map for Optimizing Observations
 • 7.6 CCD Gain Selection
 • 7.7 ACS Apertures
 • 7.8 Specifying Orientation on the Sky
 • 7.9 Parallel Observations
 • 7.10 Pointing Stability for Moving Targets
 Chapter 8: Overheads and OrbitTime Determination
 Chapter 9: ExposureTime Calculations

Chapter 10: Imaging Reference Material
 • 10.1 Introduction
 • 10.2 Using the Information in this Chapter

10.3 Throughputs and Correction Tables
 • WFC F435W
 • WFC F475W
 • WFC F502N
 • WFC F550M
 • WFC F555W
 • WFC F606W
 • WFC F625W
 • WFC F658N
 • WFC F660N
 • WFC F775W
 • WFC F814W
 • WFC F850LP
 • WFC G800L
 • WFC CLEAR
 • HRC F220W
 • HRC F250W
 • HRC F330W
 • HRC F344N
 • HRC F435W
 • HRC F475W
 • HRC F502N
 • HRC F550M
 • HRC F555W
 • HRC F606W
 • HRC F625W
 • HRC F658N
 • HRC F660N
 • HRC F775W
 • HRC F814W
 • HRC F850LP
 • HRC F892N
 • HRC G800L
 • HRC PR200L
 • HRC CLEAR
 • SBC F115LP
 • SBC F122M
 • SBC F125LP
 • SBC F140LP
 • SBC F150LP
 • SBC F165LP
 • SBC PR110L
 • SBC PR130L
 • 10.4 Geometric Distortion in ACS
 • Glossary